The pH Meter Kit includes all the essential solutions and equipment you need to take reliable pH readings of your plating solutions.
pH Meter KitSale!
You can purchase this kit as it is or if you wish, you can use it as a guide to put together your own bespoke kit. Click on the items in the list to see detailed product information.
£52.20 + VAT
Calibrating the meter with our pH 7 Buffer, in preparation for pH testing our Palladium Tank Plating Solution. To see how to use our pH Meter Kit, please click HERE for our Electroplating Manual.
Taking a pH Meter read of our Palladium Tank Plating Solution. The result from this should then be entered into our Palladium Tank Plating Calculator which can be found by clicking HERE.
Our pH Meter Kit will help you keep your tank plating solutions working at their highest efficiency. It's a good idea to monitor the pH balance of all of our tank plating solutions but our Palladium Tank Plating Solution in particular needs to have regular pH readings taken, to ensure the right amount of Palladium pH Adjuster is added.
Before you use your pH Meter, you will need to calibrate it for our solutions. You will need to use our pH 7 Buffer to do this, which is included when you buy our pH Meter. Once calibrated, the Meter can then be used to test the pH of our Palladium Tank Plating Solution. It is advisable to calibrate your pH Meter on regular basis. This guide is therefor presented in two stages; the first demonstrating how to calibrate you meter and the second showing how to take a pH reading of our Palladium Tank Plating Solution.
We recommend you use our Plating Stand with one of our Plating Arms fitted to mount the meter, as this will hold the meter safely while the reading is being taken.
1. Calibrating your pH Meter
a) Turn on your pH Meter
Turn on your Meter at the top by pushing the black switch firmly to the right.
b) Removing the Cap
Remove the cap at the end of the meter probe and squirt some tap water over the end to hydrate it, so that the meter works efficiently.
c) Pouring out the pH 7 Buffer
Pour out enough pH 7 Buffer into the small beaker to cover the end of the meter probe.
d) Mounting the Meter on the Plating Stand
Grip the black switch at the top of the Meter in the crocodile clip at the end of the Plating Arm.
What you will need
To calibrate your pH Meter and then carry out a pH test on your Palladium Tank Plating Solution you will need the following equipment and solutions;
e) Leave the Meter
Leave the Meter in the pH 7 Buffer solution for five minutes.
f) Adjuster the Meter
Adjust the readout on the screen so that it reads 7, using the yellow screwdriver to turn the small brass screw on the top of the meter marked ‘pH 7’. Your pH Meter is now calibrated.
2. Taking a pH Meter reading of our Palladium Tank Plating Solution
a) Rinsing the probe of Buffer
Take the Meter out of the crocodile clip and thoroughly rinse the Buffer from the end of the probe so that you don’t contaminate your Palladium Tank Plating Solution.
b) Testing the Palladium Tank Plating Solution
Put the Meter back in the crocodile clip on the Plating Arm and then position the end of the meter probe into the Palladium Tank Plating Solution. Leave for 5 minutes and then note down the pH. You can enter in this reading into the Palladium Tank Plating Calculator to find out how much Palladium pH Adjuster you will need to add. The optimum pH range for Palladium Tank Plating Solution is 0.7 – 0.8
Certain small items, for example Swabs, Nibs and Plating Pens can be sent by normal post within the UK
- The courier option for UK deliveries is FedEx
- Free UK mainland delivery available on orders over £100
- International orders usually 3-5 working days.
For all price quotations for deliveries, please add your items to our shopping cart. You will be able to see the shipping costs by clicking on the ‘Calculate shipping’ button in the cart and before checking out.